عنوان مقاله [English]
نویسندگان [English]چکیده [English]
The accuracy of isotopic compositions measurement is affected by all the ICP-MS instruments because of the phenomenon known as mass bias, where it is defined as difference between the measured and true values in an isotopic analysis. This research has introduced mass bias and also evaluated the effective parameters including concentration of uranium, electrical field behind the second or skimmer cone, detector voltage, plasma power and nebulizer flow. The least mass bias with a relative error -0.3% for the isotopic analysis of uranium results in aqueous media without interference of the matrix with 1.2% RSD of 0.3 mg/L concentration, while the voltages of ion lens 1 and 2 in the ICP-TOF-MS were -700V and -550V, respectively. The instrument was set in the plasma power of 1000 watts, nebulizer flow of 0.78 ml/min, and the detector voltage of -2450V.
1. M. Simon, Nelms (Ed.), ICP Mass Spectroscopy Handbook, Blackwell, CRC Press (2005) 58-62.
2. A. Montaser, Inductively Coupled Plasma Mass Spectrometry, Wiley-VCH, New York (1998) 421-501.
3. L.A. Allen, Evaluation of Isotopic Analysis, Anal Chem. 69 (1997) 2384-2391.
4. D.J. Douglas, (Ed. A. Montaser), ICP-MS, Principal and Applications, Wiley-VCH, New York (1998) 503-588.
5. T.F.D. Mason, Ph.D. Thesis, Imperial College London (2003).
6. P.H. Dawson (Ed), Analysis with ICP-MS, Elsevier, Amsterdam (1976) 143.
7. R. Thomas, Practical Guide to ICP-MS, Marcel Dekker, Maryland, USA (2004).
8. ASTM C1474, Standard Test Method for Analysis of Isotopic Composition of Uranium in Nuclear-Grade Fuel Material by Quadrupole Inductively Coupled Plasma-Mass Spectrometry (2010).
9. F. Vanhaecke, P. Degryse, Isotopic Analysis; Fundamentals and Applications Using ICP-MS, WILEY-VCH (2012).
10. H.P. Longerich, Heavy Metal Isotopic Analysis, Spectrochim Acta, 42B (1987) 39-48.
11. M. Simon, Nelms (Ed.), ICP Mass Spectroscopy Handbook, Blackwell, CRC Press (2005) 163-164.